Payment & Shipping Terms:
|Name:||ECG Similator||Application:||Medical Signal Simulate|
|Battery Type:||Buil-in Lithium|
ECG simulator 10 leads ECG machine simulator with 2 type waveform patient monitor simulator build-in battery
1.Rechargeable Lithium battery
2.10 pcs multi-functional ECG connector, can be connect with different type ECG machine cable or patient monitor cable
3.Battery power indicate light
4.Low battery power automatic turn off function, protect the Lithium battery
The simulator can generate 2 types waveform
1.Normal ECG waveform
2.Plus-minus triangular waveform
1.The simulator will automatically generate Normal ECG waveform when connect to power
2.The LED display will be off automatically if without operation after 5 seconds, press any key will be returned to display
There are 4 keys, followed by selection keys, increase key, reducing key, the Enter key, in addition to a set of key combinations
This key is used to select the parameter you want to change; there are 4 LED tubes to display the four codes represent what is displayed,
The 1st LED representative waveform code, LED 2-4 representatives the parameter (2 is the value of one hundred, 3 representatives of ten, 4 representatives of bits)
Operating method described separately for each waveform:
1. Normal ECG waveform:
Heart rate setting range: 10-200bpm; (initial value: 60 bpm)
Signal amplitude fixed.
2. Plus-minus triangular waveform:
Frequency range: 10-300bpm; (initial value: 75 bpm)
Amplitude range: 0.1-4Mv (10: 0.1mV, 400: 4mV);
2 modes, Mode 1: Positive waveform; mode II: negative waveform; Use combinations key to choose.
Note: The signal amplitude settings:
amplitude range: 0.1mV-4mV (10: 0.1mV, 400: 4mV); (initial value: 100)
Please note that this amplitude setting will affect all waveforms.
Please note that the signal amplitude settings do not affect the amplitude of the ECG waveform.